Explore our Healthcare Products & Features Online
Discover Product Specifications
Minimum PC System Requirements:
Medical System (ECG, Holter, BP) Features:
The length of the cuff is long enough that when inflated to 300mmHg, it does not slip or become loose. For hook and contact closure cuffs:
The cuff is long enough to encircle the largest circumference limb (44cm) completely. It maintains its full width throughout this length.
Pressure Capacity: The cuff completely retains the bladder when the bladder is inflated to a minimum pressure of 330mmHg.
The tubes and connectors remain intact at 330mmHg.
Cuff Closures/construction for Non-disposable Cuffs: The cuff integrity is maintained after 1000 open-close cycles and after 10,000 cycles to 300mmHg.
The cuff bladder length encircles much of the circumference of the limb throughout its length and maintains it’s much of its width throughout its length.
The bladder length is at least 0.80 times the circumference of the limb at the midpoint of the cuff application. The cuff bladder width is at least 0.37 (preferably 0.40) times the circumference of the limb at the midpoint of the cuff application.
System leakage (cuff bladder, tubing, connectors, Etc.) is less than 1 mmHg per second.
Medical Electrical Performance and Safety:
Note: the instrument's specifications may degrade if operated outside of these environmental specifications
Cassette Playback Specifications:
Holter Flash Recorder Specifications:
And we will schedule a time to take care of your needs as quickly as we can.
Thank you for your interest in Nasiff CardioCard® products.
Made in the USA
Helping Doctors save lives since 1989